The N-slit interferometer is an extension of the double-slit interferometer also known as Young's double-slit interferometer. One of the first known uses of N-slit arrays in optics was illustrated by Newton.1 In the first part of last century, Michelson2 described various cases of N-slit diffraction.
Feynman3 described thought experiments, of two-slit quantum interference, of electrons using Dirac's notation.4 This approach was extended to N-slit interferometers, by Duarte and colleagues in 1989,5 using narrow-linewidth laser illumination, that is, illumination by indistinguishable photons. The first application of the N-slit interferometer was the generation and measurement of complex interference patterns.56 These interferograms are accurately reproduced, or predicted, by the N-slit interferometric equation for either even (N = 2, 4, 6,…), or odd (N = 3, 5, 7,…), numbers of slits.6
The N-slit laser interferometer, introduced by Duarte,5610 uses prismatic beam expansion to illuminate a transmission grating, or N-slit array, and a photoelectric detector array (such as a CCD or CMOS) at the interference plane to register the interferometric signal.61011 The expanded laser beam illumination in the N-slit interferometer can also take the shape, via the introduction of a convex lens prior to the prismatic expander, of a beam extremely elongated in the propagation plane and extremely thin in the orthogonal plane.610 This use of one-dimensional (or line) illumination eliminates the need of point-by-point scanning in microscopy and microdensitometry.610 Thus, these instruments can be used as straight forward N-slit interferometers or as interferometric microscopes (see section on microscopy).
These interferometers, originally introduced for applications in imaging,6 are also useful in optical metrology and have been proposed for secure optical communications in free space,712 between spacecraft. This is due to the fact that propagating N-slit interferograms suffer catastrophic collapse from interception attempts using macroscopic optical methods such as beam splitting.7 Recent experimental developments include terrestrial intra-interferometric path lengths of 35 meters8 and 527 meters.9
These large, and very large, N-slit interferometers are also being used to study various propagation effects including microscopic disturbances on propagating interferometric signals. This work has yielded the first observation of diffraction patterns superimposed over propagating interferograms.9
These diffraction patterns (as shown in the first photograph) are generated by inserting a spider web fiber (or spider silk thread) into the propagation path of the N-slit interferogram. The position of the spider web fiber is perpendicular to the plane of propagation.9
N-slit interferometers, using large intra interferometric distances, have been found to be effective detectors of clear air turbulence.89 Here, it should be mentioned that the distortions induced by clear air turbulence, on the interferometric signal, are different in character and magnitude to the catastrophic effects resulting from attempts to optical interception of the signals using macroscopic optical elements such as beam splitters.13
As previously mentioned the original application of the N-slit laser interferometer was interferometric imaging.61014 In particular, the one dimensionally expanded laser beam (with a cross section 25-50 mm wide by 10-25 μm high) was used to illuminate imaging surfaces (such as silver-halide films) to measure the microscopic density of the illuminated surface. Hence the use of the description interferometric microdensitometer.10 Resolution down to the nano regime can be provided via the use of interinterferometric calculations.6 When used as a microdensitometer the N-slit interferometer is also known as a laser microdensitometer.14
The multiple-prism expanded laser beam is also described as an extremely elongated laser beam. The elongated dimension of the beam (25-50 mm) is in the plane of propagation while the very thin dimension (in the μm regime) of the beam is in the orthogonal plane. This was demonstrated, for imaging and microscopy applications, in 1993.610 Alternative descriptions of this type of extremely elongated illumination include the terms line illumination, linear illumination, thin light sheet illumination (in light sheet microscopy), and plane illumination (in selective plane illumination microscopy).
- Beam expander
- Clear air turbulence
- Double-slit experiment
- Free-space optical communication
- N-slit interferometric equation
- List of laser articles
- I. Newton, Opticks (Royal Society, London, 1704).
- A. A. Michelson, Studies in Optics (Chicago University, Chicago, 1927).
- R. P. Feynman, R. B. Leighton, and M. Sands, The Feynman Lectures on Physics, Vol. III (Addison Wesley, Reading, 1965).
- P. A. M. Dirac, The Principles of Quantum Mechanics, 4th Ed. (Oxford, London, 1978).
- F. J. Duarte and D. J. Paine, Quantum mechanical description of N-slit interference phenomena, in Proceedings of the International Conference on Lasers '88, R. C. Sze and F. J. Duarte (Eds.) (STS, McLean, Va, 1989) pp. 42–47.
- F. J. Duarte, On a generalized interference equation and interferometric measurements, Opt. Commun. 103, 8-14 (1993).
- F. J. Duarte, Secure interferometric communications in free space: enhanced sensitivity for propagation in the metre range, J. Opt. A: Pure Appl. Opt. 7, 73–75 (2005).
- F. J. Duarte, T. S. Taylor, A. B. Clark, and W. E. Davenport, The N-slit interferometer: an extended configuration, J. Opt. 12, 015705 (2010).
- F. J. Duarte, T. S. Taylor, A. M. Black, W. E. Davenport, and P. G. Varmette, N-slit interferometer for secure free-space optical communications: 527 m intra interferometric path length , J. Opt. 13, 035710 (2011).
- F. J. Duarte, Electro-optical interferometric microdensitometer system, US Patent 5255069 (1993).
- F. J. Duarte, in High Power Dye Lasers (Springer-Verlag, Berlin, 1991) Chapter 2.
- F. J. Duarte, Secure interferometric communications in free space, Opt. Commun. 205, 313-319 (2002).
- F. J. Duarte, Interferometric imaging, in Tunable Laser Applications, 2nd Edition (CRC, New York, 2009) Chapter 12.
- F. J. Duarte, Interferometric imaging, in Tunable Laser Applications (Marcel-Dekker, New York, 1995) Chapter 5.
- L-B. Deng, Theory of atom optics: Feynman path integral approach, Frontiers Phys. China 1, 47-53 (2006).
- H. Liu et al., Fourier-transform ghost imaging with pure far field correlated thermal light,Phys. Rev. A. 76, 053808 (2007).
- F. J. Duarte, Tunable Laser Optics (Elsevier-Academic, New York, 2003) Chapter 10.
- J. F. Clauser and J. P. Dowling, Factoring integers with Young’s N-slit interferometer, Phys. Rev. A 53, 4587–4590 (1996).